■ Ruptured Achilles tendon
■ Stable foot or ankle fractures
■ Acute ankle sprains
■ Post-op ankle surgery where protected range-of-motion is required.
■ Air can be easily adjusted to achieve desired compression and custom fit.
■ Patented ROM with orange safety lock mechanism and easy
ROM setting application.
■ ROM setting between 0°~ 40° plantar-flexion and 0°~ 40° dorsi-flexion
in 10° increments.
■ Soft-foam with breathable fabric lining keeps you dry and comfortable.
■ Fits left and right foot.
Size：S . M . L . XL
|SIZE||Max. Foot Length||US Women’s shoe||US Men’s shoe||Euro shoe|
|S||25 cm||6 – 8||4.5 – 7||36 – 40|
|M||28 cm||8.5 – 11.5||7.5 – 10.5||40 – 44|
|L||30 cm||11.5 – 13.5||10.5 – 12.5||44 – 47|
|XL||31 cm||> 13.5||> 12.5||> 47|
THE CONWELL ROM AIR SHIELD WALKER (STANDARD) is indicated following stable foot and/ or ankle fractures, ruptured Achilles tendon, post-operative procedures, where protected range-of-motion is required.
❶Untie strap and soft foam liner, position foot carefully into soft foam liner, step heel back. Steady position (Fig.A).❷Make sure the ankle is well positioned inside the walker, fasten the hook and loop fasteners of the soft foam liner (Fig.B).
❸You may put pad kits to supplement inner space or between walker frame and soft foam liner if needed (Fig.C).
❹Place front panel in correspondent place (Fig.D).
❺Fasten all the straps from ankle proximal to distal and adjust the straps to comfortable tight (Fig.E).
❻To inflate aircell, turn the button (Fig.F) towards the indicated “inflate” direction until it is tight, then press the black pump button (Fig.G) to inflate air until aircell feels snug. Stop inflating while uncomfortable compression felt.
Heavy compression may reduce blood circulation.
❼To deflate aircell, turn the button (Fig.F) towards indicated “deflate” direction until it is tight, the aircell will deflate automatically. To accelerate aircell deflating, press the black pump button (Fig.G).
ROM MECHANISM OPERATION :
❽Please push the orange buttons upwards using both hands to unlock the setting (Fig.H)
❾Press the white ROM adjustment buttons to desired ROM (Fig. I ).
❿When desired ROM is set, push the orange buttons back to white buttons’ positions to lock the ROM (Fig. J and Fig. K).
1.If discomfort or pain results, persists or increases, discontinue use and consult a physician.
2.Usage as directed by physician, also operated by professional accroding to physician indication. Do not operate without consulting physician.
3.Make sure the hook and loop fasteners are well fastened while wearing the product. If the hook and loop fasteners cannot be fastened well. stop using the product to prevent injuries.
1.Remove foam liner from hinges for hand wash. using a mild detergent and cold water. air dry. Do not wash in washing machine or place in mechanical dryer. Do not bleach nor iron.
2.For plastic components, clean with a damp cloth, air dry. Do not use detergent or bleach.